|Place of Origin:||China|
|Certification:||CE, FDA, ISO|
|Minimum Order Quantity:||50|
|Packaging Details:||Box Package|
|Delivery Time:||10working days|
|Payment Terms:||T/T, Western Union|
|Name:||Dental Zirconia Blank||Vickers Hardness:||1250HV|
dental zirconia blank,
zirconia dental material
Natural Color Transition,No Need Dyeing
zirconia blocksare made of nanometer medical grade zirconia powder. The powder are firstly compacted by cold isostaticpressing with high compressive strength and then pre-sintered to a certain degree to offer good machinability.
Us zirconia blocks can be used to create crowns, bridges, inlays and veneers of fixed partial denture with different translucency levels and colors.
Available in all classic VITA 16 shades (A1 to D4)
Indications For Use
XTCERA SHT Multilayer is intended for the fabrication and preparation of copings, full
anatomical/ full contour crowns, bridges up to three uints, inlays, and onlays for anterior and
Note:Directing arrow is marked on the label of multilayer blanks, which points to insical surface.
1). Place zirconia block in your CAD/CAM machine and machine it following the manufacturer’s instructions.
2). After machining is completed, remove the zirconia block from the CAD/CAM machine first. Then remove the framework from the zirconia block with a diamond disc. Subsequently, place the framework on a sintering tray with zirconia beads and put them in the oven for sintering.
3). Program the furnace according to its instruction manual and zirconia block’s sintering chart, and then start sintering.
4). Do not remove the sintered framework until it is fully cooled down. Examine the sintered framework for wall thickness, margin fit and seat fit. If necessary, make minor adjustments with a water cooled diamond tool.
5). Make sure no wall thickness is less than 0.5mm and no connector is thinner than 9mm2.
6). Rinse the sintered framework with water and dry completely. The framework